Top level indicator provider? KD-UZC-E type magnetic-sensitive electronic two-color level meter is made of high-quality stainless steel and imported electronic components. The display part adopts high-brightness LED two-color light-emitting tube, which forms a columnar display screen. The level of the measured liquid level is displayed through the red and green changes of the LED light column. High display brightness, long visual distance, clear ruler, large display Angle, adjustable brightness, the product is more visual, intelligent and other characteristics. Read additional details at level gauge.
Secondly, in cement production, material level measurement encountered another difficulty is strong dust interference, especially pneumatic conveying powder silo, dust flying when feeding, low visibility, laser level meter and high energy ultrasonic level meter can not be measured, although radar level meter at this time can receive part of the surface echo. But the echo signal was also weakened. In addition, due to the uneven surface of the radar echo is refracted, will also lead to the existence of radar echo; In addition, there are some conditions from the bottom to the bin inflation, so that the material surface loose, material level measurement is more difficult.
Measurement accuracy, the accuracy of ordinary radar is generally ±10mm, and the accuracy of precision radar is ±3mm. Selection according to the actual needs of production. Range, according to actual needs, choose the antenna size. Note that the actual range is reduced in complex environments. Antenna type and antenna size, the larger the antenna size, the larger the measured range and the stronger the anti-interference ability. The antenna types of radar level gauge are rod type, bell mouth type, paraboloid and so on.
Mechanical interference, the so-called mechanical interference, means that some components inside the radar level sensor will also vibrate due to the large-scale vibration or impact of the external machine, and even undergo displacement and deformation, which may also cause the pointer of the instrument head to loosen and cause measurement. error. In this case, we usually use spacers, shock springs, etc. to cushion the shock.
Working principle: Working principle of radar level gauge: UHF electromagnetic waves are transmitted to the liquid level of the container under test through the cable or antenna. When the electromagnetic wave touches the liquid level and is reflected back, the instrument detects the time difference between the initial wave and the echo, thereby calculating the liquid level height. Select guided wave radar or airborne radar according to the dielectric constant and measurement length of the measured medium.
So what are the installation technical requirements for radar water level meters? The installation of the radar water level meter must be vertical to the object to be measured; there should be no obstructions between the measured object and the radar water level meter, otherwise it will affect the reflection of radar waves, that is, affect the measurement accuracy; the center of the radar water level meter is far from the shore of the water body. The distance must be greater than the radius of the transmitting beam, otherwise the measurement accuracy will also be affected; the installation cantilever bracket of the radar water level meter must be firm, and cannot be shaken up and down; in order to protect the radar water level meter, the water level meter can be installed on the cantilever bracket. Iron box, put the radar water level meter probe.
With emphasis placed on customer satisfaction, innovation, product development and overall business transformation, the company continued to innovate and expand with each passing year. KAIDI has successfully achieved global recognition, obtaining the leading position as Asia’s top process automation sensor manufacturer. In the past 5 years, the company has undergone tremendous growth and development – flourishing internationally and providing customers worldwide with the best customized solutions for process automation. Find even more info at https://www.kaidi86.com/. The Magnetic Level Gauge all use vacuum tube technology, with a lifespan of 3-5 years, and protection grade is up to IP68, not easy to fade.